Questions and Answers
Can other plates be used with the MVS System?
Yes, however traceability of the results will only be maintained if MVS Verification Plates are used. Additional library plates complete with manufacturer’s defined well dimensions and product numbers are listed in the Plate Library.
Can the Verification Plates be washed and reused?
No. The MVS is specifically manufactured and optimized to minimize error. Reusing a Verification Plate potentially introduces a source for error.
Can a Verification Plate be re-read?
Yes. Once a Verification Plate scan has completed, the filled wells in the plate can be re-read within 60 minutes** provided justification is entered.After 60 minutes, the wells that were read are locked out and cannot be re-read. Empty wells not already read can be read using a different layout in any future verification.
**applicable in Data Manager version 3.1 and above. Data Manager 3.0 and below have a 20 minute lockout period.READ MORE +
**applicable in Data Manager version 3.1 and above. Data Manager 3.0 and below have a 20 minute lockout period.